One point calibration
Sometimes a one point calibration is only needed. Typically, a one-point calibration is made by placing the electrode in pH = 7.00 buffer and adjusting the meter to read 7.00, ZERO (offset). The Slope is kept at 100% or at the previously calibrated value. This method gives a reasonable accuracy.
One-point calibration is also known as Offset-only calibration.