Keep in mind that all pH probes age with time, even if it is not in use.
• The decomposition process
that occurs when glass surfaces are in contact with aqueous solutions,
will change the chemical composition of the glass membrane.
• As aging progresses, the hydrated gel layer becomes thicker, resulting in an increase in the gel layer resistance.
• Damage and degradation of the hydrated gel layer due to chemical or mechanical cleaning and demanding measurement conditions.
• Response will become slower (increase in gel layer resistance).
• During calibration, you will observe an increasing zero point offset and a declining slope value.
• As the probe aging process continues you will need to calibrate more frequently.
• Finally, erratic readings. Now it's definitely time to replace your pH sensor.
• incorrect handling of the pH sensor when not in use. Proper storage procedure is important.
• measuring strong acids and strong alkaline solutions. Strong acids and, to a greater extent, strong alkaline solutions attack the pH sensitive glass membrane. Even neutral solutions containing high concentrations of alkali ions, and sodium ions attack the glass membrane.
• using in high-temperature environments (> 60°C or 140°F). Increasing temperature will increase the rate of the chemical attack.
Take the opportunity to share your knowledge and experiences with others.
Please, use the form below to publish your contribution about the subject "Aging of pH probes". Before your page is published, it will be reviewed to make sure it applies to the submission guidelines for this website.
Don't forget to read what others have written, maybe you have a comment to add.
Click below to see contributions from other visitors to this page...